TY - JOUR

T1 - Using the Monte Carlo stochastic method to determine the optimal maintenance frequency of medical devices in real contexts

AU - Miguel Cruz, Antonio

AU - Aya Parra, Pedro Antonio

AU - Ocampo, Andres Felipe Camelo

AU - Guao, Viena Sofia Plata

AU - Correal O., Hector H.

AU - Hernández, Nidia Patricia Córdoba

AU - Cruz, Angelmiro Núñez

AU - Rojas, Jefferson Steven Sarmiento

AU - Torres, Daniel Alejandro Quiroga

AU - Rodríguez-Dueñas, William Ricardo

PY - 2019/1/1

Y1 - 2019/1/1

N2 - The purpose of this study was to implement and validate a Monte Carlo Algorithm (MCA) to determine the best T value (the time between two preventative maintenances) that optimizes the achieved availability of equipment types. In doing so, we (1) collected 796 maintenance works orders from 16 medical devices installed in a 900-bed hospital; (2) we fitted the probability distributions for each of the inputs of the achieved availability mathematical model (the mean preventative and corrective service time (in hours)); (3) we generated a set of random inputs following a Weibull distribution of the achieved availability mathematical model; (4) we calculated the achieved availability for every random input generated; this process was repeated for “m” iterations (an accuracy of 1%, 95% CI, alpha = 0.05); (5) the trends of the mean achieved availability for the different maintenance T intervals versus mean time to failure (MTTF) for all the equipment types were plotted; finally, (6) the best T value with the maximum value of the achieved availability of a medical device type for a specific MTTF was the optimal target. The mean simulation time for all the cases was 12 min. The MCA was able to determine the best T value, optimizing the achieved availability in 81.25% of cases. In conclusion, the results showed that, on average, the T maintenance intervals determined by the MCA were statistically significantly different from the original T values suggested either by the clinical engineering department or third-party maintenance providers (MCATmean = 1.68 times/yr, ActualTmean = 2.56 times/yr, p = 0.008).

AB - The purpose of this study was to implement and validate a Monte Carlo Algorithm (MCA) to determine the best T value (the time between two preventative maintenances) that optimizes the achieved availability of equipment types. In doing so, we (1) collected 796 maintenance works orders from 16 medical devices installed in a 900-bed hospital; (2) we fitted the probability distributions for each of the inputs of the achieved availability mathematical model (the mean preventative and corrective service time (in hours)); (3) we generated a set of random inputs following a Weibull distribution of the achieved availability mathematical model; (4) we calculated the achieved availability for every random input generated; this process was repeated for “m” iterations (an accuracy of 1%, 95% CI, alpha = 0.05); (5) the trends of the mean achieved availability for the different maintenance T intervals versus mean time to failure (MTTF) for all the equipment types were plotted; finally, (6) the best T value with the maximum value of the achieved availability of a medical device type for a specific MTTF was the optimal target. The mean simulation time for all the cases was 12 min. The MCA was able to determine the best T value, optimizing the achieved availability in 81.25% of cases. In conclusion, the results showed that, on average, the T maintenance intervals determined by the MCA were statistically significantly different from the original T values suggested either by the clinical engineering department or third-party maintenance providers (MCATmean = 1.68 times/yr, ActualTmean = 2.56 times/yr, p = 0.008).

UR - http://www.scopus.com/inward/record.url?scp=85048303453&partnerID=8YFLogxK

UR - http://www.scopus.com/inward/citedby.url?scp=85048303453&partnerID=8YFLogxK

U2 - 10.1007/978-981-10-9023-3_49

DO - 10.1007/978-981-10-9023-3_49

M3 - Conference article

AN - SCOPUS:85048303453

VL - 68

SP - 273

EP - 277

JO - IFMBE Proceedings

JF - IFMBE Proceedings

SN - 1680-0737

IS - 3

T2 - World Congress on Medical Physics and Biomedical Engineering, WC 2018

Y2 - 3 June 2018 through 8 June 2018

ER -